Mid-tracheal, postintubation stenosis can be managed with an extended length tracheostomy tube to bypass the stenotic area. However these extra-long tracheostomy tubes are not fenestrated, and when the stenotic tracheal lumen sits against the tracheostomy tube, phonation is not possible as there is no translaryngeal airflow. A 59-year-old man developed distal tracheal stenosis following a prolonged intubation and tracheostomy after a motorcycle accident. He eventually required an extra-long tracheostomy tube to bypass the stenotic region. We modified a silicone tracheostomy tube by creating a fenestration on its posterior wall. This relieved the obstruction while still allowing phonation and speech.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.