Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.
A 42-year-old woman was scheduled for a laparoscopic cholecystectomy. She was American Society of Anesthesiologists grade 1 and had no obvious anomalies of the airway, Mallampatti score was 2.
After induction, laryngoscopy revealed a Cormack and Lehane grade II view. Intubation was with a size 8.0 cuffed tracheal tube (TT). A lubricated 22F, 78 cm orogastric tube (OGT) was then inserted blindly with minimal resistance to 55 cm. Suction was applied, producing a small amount of thin yellow secretions; a bile bag was then attached.
The surgeons commented that the stomach was adequately deflated however no gastric contents were evident in the bile bag.
At the end of the operation the OGT was suctioned and initially met no resistance upon removal. At half way mild resistance was felt. At approximately 20 cm there was a sudden increase in resistance, even gentle traction on the OGT created movement of the TT. We decided to leave the partially removed OGT tube in place as no ventilatory compromise was evident.
The patient was extubated, removing what is shown in figure 1.
We theorise that the OGT looped around the TT on insertion and formed a loose knot in the mouth that did not impede advancement into the oesophagus. Upon removal, the tension applied caused the knot to be drawn further down the OGT and tighten around the TT.
There have been reports of gastric tubes becoming knotted around the TT.1–3 There is a common theme in each case: all were passed blind and had difficulty in placing the gastric tube.
Even gastric tubes that meet no or minimal resistance when passed blind have the potential for knotting or misplacement.
Place gastric tubes under direct vision to avoid looping in oropharynx in anaesthetised patients.
If any resistance is felt upon removal of gastric tube have a high index of suspicion for knotting to the TT because continued attempt at extraction could cause unintentional extubation3 with respiratory compromise or larnygospasm.
Patient consent Obtained.
Competing interests None.
Provenance and peer review Not commissioned; externally peer reviewed.