Distal radius fractures are one of the most common upper extremity injuries seen by hand surgeons each year. Many of these fractures require reduction and surgical fixation because of displacement, comminution or inherent fracture instability. New hardware is designed and introduced each year to help surgeons manage these injuries. We report a case of a major complication from the use of the Dorsal Nail Plate hardware. The patient presented to our clinic with an extensor pollicis longus attrition rupture, which required tendon transfer for treatment. Intraoperatively, we also found ongoing extensor digitorum communis tendon attrition, myostatic atrophy of the extensor pollicus longus and ultimately a large bony defect following complex hardware removal. This case serves as a reminder to consider the potential complication profile of any hardware or technique you are considering using in your patients and to evaluate donor muscles and tendons when performing grafting or transfers.
- orthopaedic and trauma surgery
- plastic and reconstructive surgery
Statistics from Altmetric.com
Contributors DR-F was involved in information acquisition, case report planning, background review, case report writing, discussion development, formatting, submission and editing of the case report. AS was involved in information acquisition, case report planning, case report writing, case report editing, discussion development and editing of the case report.
Funding The authors have not declared a specific grant for this research from any funding agency in the public, commercial or not-for-profit sectors.
Competing interests None declared.
Patient consent for publication Obtained.
Provenance and peer review Not commissioned; externally peer reviewed.
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.